Panther’s Quartz Probe is a highly precise, down hole electronic pressure/temperature memory probe. The Panther Probe is designed with the leading technology that is unmatched by any other product currently available.
Panther Probe uses Quartzdyne technology that provides our gauge with high resolution, reproducibility, and long-term stability combined with the ruggedness needed for successful long term well testing.
Our Quartz probe comes assembled, pre-programmed and ready to go. The software provided is easy to use for changing programs and/or downloading data.
Panther Quartz Probe is designed for long term testing and is very cost efficient to operate.
Download Complete Brochure
|Accuracy (% of full scale)
|Resolution (% of full scale)
||0.00006 or 0.02 kPa 35000 kPa gauge
|Repeatability (% of full scale)
||< 0.01 error
||3.5 mPa - 138 mPa (5,000 psi - 20,000 psi)
||0.2 ° C
||0.01 ° C
||160 ° C
|Number of Data Sets
||3,000,000 sets expandable to 4,000,000
|Battery Capacity (days)
|73 @ 5 second | 209 @ 60 seconds
|1.32 inch (3.35 cm) | 1.25 inch (3.18 cm)
||22.2 inch (56.5 cm) | 26.6 inch (67.5 cm)
||Quartzdyne quartz resonator
||Hastelloy C-276 and Incolloy 718 / 925
||1 CC Pack